Instruction/ maintenance manual of the product BP791IT Omron Healthcare
Go to page of 60
10 series Plus Blood Pressur e Monitor with C omF it TM Cuff Model BP791IT INSTRUCTION MANU AL ESP AÑOL ENGLISH.
2 Before Using the Monitor Introduction . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .4 Safety Information . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .5 Operating the Device .
T aking a Measurement . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .30 Calibration Check System . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . . .30 Cuff W rap Guide . . . . . . . . . . . . . . . . . . .
4 Thank you for purchasing the Omron ® BP791IT 10 series Plus Automatic Blood Pressure Monitor with ComFit TM Cuff. INTRODUCTION Fill in for futur e reference . DA TE PURCHASED: __________________ SERIAL NUMBER: ____________________ Y our new blood pressure monitor uses the oscillometric method of blood pressure measurement.
5 T o assur e the corr ect use of the pr oduct basic safety measur es should always be followed including the warnings and cautions listed in this instruction manual. W ARNING Indicates a potentially hazardous situation which, if not avoided, could result in death or serious injury .
OPERA TING THE DEVICE (continued) This device is intended for use in measuring blood pressure and pulse rate in the adult population. Do not use this device on infants or persons who cannot express their intentions. Read the Special Conditions section (page 37) of the instruction manual if your systolic pressure is known to be more than 220 mmHg.
RISK OF ELECTRICAL SHOCK Do not plug or unplug the power cord into the electrical outlet with wet hands. Do not overload power outlets. Plug the device into the appropriate voltage outlet. CARE AND MAINTENANCE Do not subject the monitor to strong shocks, such as dropping the unit on the floor .
8 BEFORE T AKING A MEASUREMENT T o ensure a reliable reading follow these recommendations: 1. A void eating, drinking alcohol, smoking, exercising, and bathing for 30 minutes prior to taking a measurement. Rest for at least 15 minutes prior to taking the measurement.
9 KNOW Y OUR UNIT Main Unit: Air Jack Battery Compar tment Start/Stop Button USER ID Selection Switch Morning/Evening A verage ( ) Button Memor y ( ) Button Display AC Adapt er Jack Operating Instruct.
KNOW Y OUR UNIT 10 Componen ts: Arm Cuff Air T ube Air Plug (Connects to Air Jack) AC A dapter Storage Case Housing Cord 10 series Blood Pressure Monitor with ComFit TM Cuff Model BP791IT INSTRUCTION .
11 UNIT DISPLA Y Diastolic Blood Pressure Systolic Blood Pr essure Movement Symbol Pulse Display / Memory Number Date/Time Display Low Battery Indicator Deflation Symbol Morning Av erage Symbol Heartb.
12 IRREGULAR HEAR TBEA T SYMBOL ( ) When the monitor detects an irregular rhythm two or more times during the measurement, the Irregular Heartbeat Symbol ( ) will appear on the display with the measurement values.
Internationally recognized guidelines for upper limit of normal home blood pressure readings are as follows: Home Guideline for Upper Limit of Normal BP Systolic Blood Pressure 135 mmHg Diastolic Blood Pressure 85 mmHg This is a general guideline because blood pressure depends on age and morbidity .
HEAR TBEA T SYMBOL ( ) The Heartbeat symbol flashes on the display at every heartbeat during the measurement. A VERAGE V ALUE SYMBOL ( ) The A verage V alue Symbol is displayed when you press the memory button. The most recent average reading appears on the display screen.
IMPORT ANT SAFET Y INSTRUCTIONS SA VE THESE INSTRUCTIONS NO TES: • Use only the authorized Omron AC Adapter that came with this monitor . • W e recommend you install the batteries even if the AC Adapter is used. If no batteries are installed, you may need to reset the date and time if the AC Adapter is disconnected.
TO CONNECT THE AC ADAPTER ① Insert the AC Adapter Plug into the AC Adapter Jack on the back side of the monitor as illustrated. ② Plug the AC Adapter into a 120V AC outlet (60 cycles). TO DISCONNECT THE AC ADAPTER ① Hold the Housing to remove the AC Adapter from the electrical outlet.
1. Remove the battery cover by pulling the hook up while pushing it in the direction of the arrow . 2. Install 4 “AA” size batteries so the + (positive) and - (negative) polarities match the polarities of the battery compartment as indicated. 3. Replace the battery cover .
BA TTERY REPLACEMENT When the Low Battery Indicator appears on the display screen turn the monitor off and remove all the batteries. Replace with four new batteries at the same time. Longlife alkaline batteries are recommended. W ARNING If battery fluid should get in your eyes, immediately rinse with plenty of clean water .
19 Set the monitor to the current date and time before taking a measurement for the first time. If the date and time are not set correctly , the measurement values stored in the memory and the average values may not be correct. If the date and time are not set, “-:--” appears during or after measurement.
20 SET TING THE DA TE AND TIME 1. TO ST AR T THE SETTING Press and hold the SET button to adjust the date and time. The year flashes on the display . 2. SETTING THE YEAR The year can be set between 2010 and 2040. When the display reaches 2040, it will return to 2010.
3. SETTING THE MONTH Press the Up button ( ) to advance by one month. Press the SET button to set the current month. The day flashes on the display . 4. SETTING THE DA Y Press the Up button ( ) to advance by one day . Press the SET button to set the current day .
22 5. SETTING THE HOUR The time is set using AM or PM. Press the Up button ( ) to advance by one hour . Press the SET button to set the current hour . The minute flashes on the display . 6. SETTING THE MINUTE Press the Up button ( ) to advance by one minute.
The T ruRead TM Mode takes three consecutive measurements. The monitor will inflate, take a measurement, and deflate three times separated by a short interval of time between each measurement. 1. Select your USER ID (A or B). 2. Press the SET button. The T ruRead TM symbol and the oFF setting appears on the display .
4. Press the SET button to confirm the change. If on is selected, the number of seconds between each measurement is displayed. NO TE: The default value between each measurement is 60 seconds. 5. Press the Up or Down button ( / ) to change the interval of time between each measurement.
7. If the date and time are correct, press the ST AR T/STOP button to turn the monitor off. T o change the date and time, refer to “Setting the Date and T ime”.
CAUTION Read all of the information in the instruction manual and any other literature in the box before operating the unit. CAUTION This device is intended for use in measuring blood pressure and pulse rate in the adult population. Do not use this device on infants or persons who cannot express their intentions.
27 3. Sit on a chair with your feet flat on the floor . Place your left arm on a table so the cuff is level with your heart. 4. Hold the thumb grip on the cuff securely with your right hand.
28 6. Apply the cuff to your left upper arm so the blue stripe is on the inside of your arm and aligned with your middle finger . The air tube runs down the inside of your arm. The bottom of the cuf f should be approximately 1/2" above your elbow .
APPL YING THE CUFF ON THE RIGHT ARM When taking a measurement using the right arm use this instruction for Step 6 on page 28. Apply the cuff to your right upper arm so the thumb grip is centered on the inside of your inner arm. The bottom of the cuf f should be approximately 1/2'' above your elbow .
The monitor is designed to take measurements and store the measurement values in the memory for two people using USER ID A and USER ID B. The monitor can also be used to take a single measurement for other persons using the GUEST Mode. The USER ID symbol is not displayed when using the GUEST Mode.
USING THE GUEST MODE 1. Press and hold the ST AR T/STOP button. The A and B Symbols disappear from the display . 2. Release the ST AR T/STOP button. The cuff starts to inflate automatically . NO TE: T o stop the inflation press the ST AR T/STOP button.
SELECTING THE USER ID Select your USER ID. Always use the same USER ID when taking a measurement. The monitor stores the measurement values in the memory by the USER ID selected. These measurement values are used to calculate the Morning A verages and Evening A verages.
USING THE SINGLE MODE 1. Select your USER ID (A or B). 2. Press the ST AR T/STOP button. All symbols appear on the display . The USER ID Symbol you selected flashes on the display . The cuff starts to inflate automatically . As the cuff inflates, the monitor automatically determines your ideal inflation level.
3. Inflation stops automatically and the measurement is started. As the cuff deflates, decreasing numbers appear on the display . The Heartbeat Symbol ( ) flashes at every heartbeat. 4. When the measurement is complete, the arm cuf f completely deflates.
USING THE TRUREAD TM MODE 1. Select your USER ID (A or B). 2. Press the ST AR T/STOP button. All symbols appear on the display . The USER ID Symbol you selected flashes on the display . The cuf f starts to inflate automatically . NO TE: T o stop the inflation or measurement, press the ST AR T/STOP button.
T AKING A MEASUREMENT 36 The second measurement is complete, waiting to start the next measurement. The consecutive measurements are complete. The average for three measurements appears on the display . 5. Press the SET button to display the blood pressure and pulse rate for the first measurement.
T AKING A MEASUREMENT 37 If your systolic pressure is known to be more than 220 mmHg, press and hold the ST AR T/STOP button until the monitor inflates 30 to 40 mmHg higher than your expected systolic pressure. NO TES: • Do not apply more pressure than necessary .
The monitor is designed to store the blood pressure and the pulse rate in the memory for two people (USER A and USER B) every time a measurement is completed. The monitor automatically stores up to 100 sets of measurement values (blood pressure and pulse rate) for each user (A and B).
TO DISPLA Y THE MEASUREMENT V ALUES 1. Select your USER ID (A or B). 2. Press the Memory button ( ). The A verage Symbol ( ) appears on the display when the average reading is shown.
3. Press the Down button ( ) to display the most recent set of measurement values on the screen. The values are displayed from the most recent to the oldest. NO TE: The date and time are alternately displayed with the measurement values. Press the Up or Down button to display the next set of values.
MORNING AND EVENING A VERAGES The monitor calculates and displays a weekly average for measurements taken in the morning ( ) and evening ( ). The monitor stores 8 weeks of Morning A verages and 8 weeks of Evening A verages for each user (A and B). NO TE: The week begins Sunday at 4:00 AM.
TO DISPLA Y MORNING AND EVENING A VERAGES 1. Select your USER ID (A or B). 2. Press the Morning/ Evening A verage button ( ) to select the Morning or Evening average. The average for the current week “THIS WEEK” appears on the display . 3. Press the Down ( ) button to display the previous weeks.
43 USING THE MEMORY FUNCTION DISPLA Y COMBINA TIONS In addition to the Morning A verage and Evening A verage Symbols, the monitor may also display the Morning Hypertension Symbol if the morning average for that week is above the internationally recognized guidelines.
44 The Morning Hypertension Symbol () appears if the weekly average for morning measurements is above 135/85. In this case the Morning Hypertension Symbol () is displayed when the Evening A verage is displayed, regardless of the values for the Evening A verage.
Optional Instructions for PC Connection T o install software, choose either Microsoft ® HealthV ault TM or Omron Health Management software. NO TE: Any measurement values taken before setting the date and time are not transferred to Microsoft HealthV ault TM and Omron Health Management software.
OPERA TING ENVIRONMENT REQUIREMENTS The following operating environments are required to run Microsoft ® HealthV ault. Supported OS * : HealthV ault works best with the following operating systems: E.
DOWNLOADING AND INST ALLING OMRON HEAL TH MANAGEMENT SOFTW ARE Omron Health Management Software allows you to view , process and print data about blood pressure and walking measured by Omron’ s compatible blood pressure monitor and pedometer . • T o install Omron Health Management Software onto your PC, you must have administrative privileges.
OPERA TING ENVIRONMENT REQUIREMENTS The following operating environments are required to run the Omron Health Management software. Supported OS: Microsoft ® W indows ® 2000 Professional (SP4) Micros.
For help using the software, double-click [Omron Health Management Software Users Manual.pdf] shortcut on desktop, or click [Help] in this software after starting up. 1. Double-click the [Omron Health Management Software] shortcut icon on the desktop.
3. In the New User Registration box, enter your name (e.g., Chuck) and click [OK]. NO TES: • Omron Health Management Software can manage multiple users’ data. Data is controlled by user name. • The number of users who can register for this software varies depending on the amount of free disc space in your PC.
Use only the authorized Omron USB cable, Model HHX-CABLE- USB2, included with this unit. CAUTION Use only Omron authorized parts and accessories. Parts and accessories not approved for use with the device may damage the unit. 1. Open the USB Port co ver .
4. Click [Download Data] icon to import the data. 5. The C onfirm dialog appears. Make sure y our cable is connected then click [ Y es]. 6. Select the blood pressure monitor being used and click [OK]. 7. Start impor ting data. Click [Cancel] if y ou want to cancel.
53 T o keep your blood pr essur e monitor in the best condition and pr otect the unit fr om damage, follow the directions listed below: Keep the monitor in the storage case when not in use. Make sure the AC adapter is placed under the main unit so that it does not damage the display .
Use the unit consistent with the instruction provided in this manual. CAUTION Changes or modification not approved by Omron Healthcare will void the user warranty . Do not disassemble or attempt to repair the unit or components. CAUTION Use only Omron authorized parts and accessories.
Contact Customer Service for repair information. Sensor may be out of alignment. Refer to “Applying the Arm Cuff”. Cuff is not applied correctly . Air plug disconnected. Insert the plug securely . Refer to “Applying the Arm Cuff”. Arm cuff not applied correctly .
56 ERROR INDIC A TORS AND TROUBLESHOO TING TIPS PROBLEM CA USES AND SOLUTIONS No power . No display appears on the unit. Replace all four batteries with new ones. Check the battery installation for proper placement of the battery polarities. Measurement values appear too high or too low .
57 FC C ST A TEMENT NO TE: POTENTIAL FOR RADIO/TELEVISION INTERFERENCE (for U.S.A. only) This product has been tested and found to comply with the limits for a Class B digital device, pursuant to part 15 of the FCC rules. These limits are designed to provide reasonable protection against harmful interference in a residential installation.
LIMITED W ARR ANT Y 58 Y our Omr on ® BP791IT 10 series Plus Automatic Blood Pressure Monitor , ex cl ud in g the arm cuff and batteries, is warranted to be free from defects in materials an d w or km ans hip appearing within 5 years from the date of purchase, when used in accordance with the instructions provided with the monitor .
59 Model: BP791IT (HEM-7222-ITZ) Display: LCD Digital Display Measurement Range: Pressure: 0 to 299 mmHg, Pulse: 40 to 180/min. Accuracy/Calibration: Pressure: ±3mmHg or 2% of reading Pulse: ±5% of .
Manufactured by OMRON HEAL THCARE Co., Ltd. Made in China Distributed by: Omron Healthcar e, Inc. 1200 Lakeside Drive Bannockburn, IL 60015 www .omronhealthcare.
An important point after buying a device Omron Healthcare BP791IT (or even before the purchase) is to read its user manual. We should do this for several simple reasons:
If you have not bought Omron Healthcare BP791IT yet, this is a good time to familiarize yourself with the basic data on the product. First of all view first pages of the manual, you can find above. You should find there the most important technical data Omron Healthcare BP791IT - thus you can check whether the hardware meets your expectations. When delving into next pages of the user manual, Omron Healthcare BP791IT you will learn all the available features of the product, as well as information on its operation. The information that you get Omron Healthcare BP791IT will certainly help you make a decision on the purchase.
If you already are a holder of Omron Healthcare BP791IT, but have not read the manual yet, you should do it for the reasons described above. You will learn then if you properly used the available features, and whether you have not made any mistakes, which can shorten the lifetime Omron Healthcare BP791IT.
However, one of the most important roles played by the user manual is to help in solving problems with Omron Healthcare BP791IT. Almost always you will find there Troubleshooting, which are the most frequently occurring failures and malfunctions of the device Omron Healthcare BP791IT along with tips on how to solve them. Even if you fail to solve the problem, the manual will show you a further procedure – contact to the customer service center or the nearest service center