Instruction/ maintenance manual of the product BP652 Omron Healthcare
Go to page of 30
ESP AÑOL ENGLISH 7 series A utomatic W rist Blood Pr essure Monitor Model BP652 INSTRUCTION MANU AL.
2 TABLE OF CONTENTS Before using the Monitor Introduction ......................................................................................................... 3 Safety Information ..................................................................
3 INTRODUCTION Thank you for purchasing the OMRON ® BP652 IntelliSense ® Automatic W rist Blood Pressure Monitor . The monitor is compact and easy-to-use at home, at work, and during travel. Perfect for people who frequently monitor their own blood pressure.
4 SAFETY INFORMA TION T o assure the corr ect use of the product, basic safety measur es should always be followed including the warnings and cautions listed in this instruction manual. SAFETY SYMBOLS USED IN THIS INSTRUCTION MANUAL WARNING Indicates a potentially hazardous situation which, if not avoided, could result in death or serious injury .
5 SAFETY INFORMA TION Do not use a cellular phone near the device. It may result in an operational failure. Use only Omron authorized parts and accessories. Parts and accessories not approved for use with the device may damage the unit. Useonly1.
6 BEFORE T AKING A MEASUREMENT T o ensure a reliable reading follow these recommendations: 1. A void eating, drinking alcohol, smoking, exercising, and bathing for 30 minutes before taking a measurement. Rest for at least 15 minutes before taking the measurement.
7 KNOW Y OUR UNIT Main Unit: Components: 2 “ AAA ” Alkaline Batteries Instruction Manual Quick Start Guide Date/Time setting ( ) button Display Wrist cuff Memory (MEM) button ST ART/STOP butt on U.
8 UNIT DISPLA Y Positioning indicat or Diastolic blood pressure Irregular heartbeat symbol Memory symbol (Displayed when viewing values stored in memory) A verage v alue symbol (Refer to “Using the Memory Function” .
9 DISPLA Y SYMBOLS IRREGULAR HEAR TBEA T SYMBOL ( ) When the monitor detects an irregular rhythm two or more times during the measurement, the Irregular Heartbeat Symbol ( ) will appear on the display with the measurement values.
10 This is a general guideline because blood pressure depends on age and morbidity . A lower home blood pressure goal may be recommended for certain patients, including diabetic patients, pregnant women, and patients with renal failure. WARNING Contact your physicianfor specicinformation aboutyour blood pressure.
11 BA T TERY INST ALLA TION CAUTION Useonly1.5V“AAA”alkalinebatterieswiththisdevice. 1. Press the hook on the bottom of the battery cover and lift the cover off in the direction of the arrow .
12 BATTERY REPLACEMENT Low Battery Indicator When the Low Battery Indicator appears on the display , turn the monitor off and remove all the batteries.
13 SET TING THE DA TE AND TIME Setthemonitortothecurrentdateandtimebeforetakingameasurementfortherst time. 1. TO ST ART THE SETTING Press the Date/T ime Setting button ( ) to adjust the date and time.
14 4. SETTING THE DA Y Press the Date/T ime Setting button ( ) to advance by increments of one day . Press the Date/T ime Setting button ( ) to set the current day . Thehourashesonthedisplay . 5. SETTING THE HOUR The time is set using AM or PM.
15 SET TING THE ADV ANCED POSITIONING SENSOR (APS) Y ou can change the options for various settings on your monitor . Pressing the SET button to change a setting.
16 SETTING FOR RIGHT OR LEFT WRIST MEASUREMENT 1. While in the setting mode, press the SET button until the wrist symbol ( or ) appears on the display . NOTES: •Thedefaultsettingis“ ”. •Afteryouchoose“oFF”forthe Positioning indicator , the monitor skips this process.
17 IfthePositioningindicatoris“on1”or“on2”: on1: The buzzer sounds only when the monitor is in proper position. on2: The buzzer sounds a series of two short blips if your wrist is too far away from the measuring position, and beeps when your wrist is in the proper position.
18 CAUTION Read all of the information in the instruction manual and any other literature in the box before operating the unit. CAUTION This device is intended for use in measuring blood pressure and pulse rate in the adult population. Do not use this device on infants or persons who cannot express their intentions.
19 5. Fold the remaining part of the wrist cuff back out of the way . APPL YING THE CUFF ON THE RIGHT WRIST When taking a measurement using the right wrist position the cuff as shown in the illustration. Referto“SettingforRightorLeftW ristforMeasurement”.
20 The monitor is designed to take measurements and store the measurement values in memory for two people using USER ID A and USER ID B. The monitor can be used to takeasinglemeasurementusingtheGUESTmodearenotstoredinthememory .
21 T AKING A MEASUREMENT 5. Press the ST AR T/STOP button. All symbols appear on the display . Thecuffstartstoinateautomatically . Asthecuffinates,themonitorautomaticallydeterminesyouridealinationlevel.
22 USING THE MEMOR Y FUNCTION Every time you complete a measurement the monitor stores the blood pressure and pulse rate for each user (A and B) in the memory . The monitor automatically stores up to 100 sets of measurement values (blood pressure and pulse rate) for each user (A and B).
23 USING THE MEMOR Y FUNCTION NOTES: • Thedateandtimearealternatelydisplayedwiththemeasurement values. •PresstheMEMbuttonrepeatedlytodisplaythenextvalues. •PressandholdtheMEMbuttontodisplaythevaluesfaster .
24 CARE AND MAINTENANCE T o keep your digital blood pressur e monitor in the best condition and protect the unit fr om damage, follow the directions listed below: Do not forcefully bend the cuf f. Do not fold tightly . Clean the monitor with a soft dry cloth.
25 ERROR INDICA TORS SYMBOL CAUSE CORRECTION Monitor could not detect pulse wave. T ake another measurement and remain still until the measurement is complete. Cuffunder -inated. Cuff not applied correctly . Remove the monitor from your wrist. Read“T akingaMeasurement”.
26 TROUBLESHOO TING TIPS PROBLEM CAUSES AND SOLUTIONS No power . No display appears on the unit. Replace both batteries with new ones at the same time. Check the battery installation for proper placement of the battery polarities. Measurement values appear too high or too low .
27 FC C ST A TEMENT Note: POTENTIAL FOR RADIO/TELEVISION INTERFERENCE (for U.S.A. only) This product has been tested and found to comply with the limits for a Class B digital device, pursuant to part 15 of the FCC rules. These limits are designed to provide reasonable protection against harmful interference in a residential installation.
28 Y our BP652 IntelliSense ® Automatic W rist Blood Pressure Monitor , excluding the wrist cuff and batteries, is warranted to be free from defects in materials and workmanship appearing within 5 years from the date of purchase, when used in accordance with the instructions provided with the monitor .
29 SPECIFICA TIONS Model: BP652 (HEM-6052-Z) Display: LCD Digital Display Measurement Range: Pressure: 0 to 299 mmHg, Pulse: 40 to 180/min. Accuracy/Calibration: Pressure: ±3mmHg or 2% of reading Pul.
Omron Healthcar e, Inc. 1200 Lakeside Drive Bannockburn, IL 60015 www .omronhealthcare.com Copyright © 2010 Omron Healthcare, Inc. Made in China 5329415-9A.
An important point after buying a device Omron Healthcare BP652 (or even before the purchase) is to read its user manual. We should do this for several simple reasons:
If you have not bought Omron Healthcare BP652 yet, this is a good time to familiarize yourself with the basic data on the product. First of all view first pages of the manual, you can find above. You should find there the most important technical data Omron Healthcare BP652 - thus you can check whether the hardware meets your expectations. When delving into next pages of the user manual, Omron Healthcare BP652 you will learn all the available features of the product, as well as information on its operation. The information that you get Omron Healthcare BP652 will certainly help you make a decision on the purchase.
If you already are a holder of Omron Healthcare BP652, but have not read the manual yet, you should do it for the reasons described above. You will learn then if you properly used the available features, and whether you have not made any mistakes, which can shorten the lifetime Omron Healthcare BP652.
However, one of the most important roles played by the user manual is to help in solving problems with Omron Healthcare BP652. Almost always you will find there Troubleshooting, which are the most frequently occurring failures and malfunctions of the device Omron Healthcare BP652 along with tips on how to solve them. Even if you fail to solve the problem, the manual will show you a further procedure – contact to the customer service center or the nearest service center