Instruction/ maintenance manual of the product 4800 Tanita
Go to page of 9
Z E R O P O U N D S K I L O S O N O F F Pounds Kilos w w w . t a n i t a . c o m Push “ON” button and wait until zero appears before stepping on scale WEIGHT Model 4800 Wheelchair Scale Cautions and W arnings ......................................
T anita Cor poration’ s Pro Plus line of electronic scales uses the most sophisticated microprocessor technology av ailab le. Each precision instrument is designed to provide accurate , reliable and repeatab le weight measurements, as w ell as many other f eatures that mak e the weighing process simple, f ast and con v enient.
ASSEMBL Y INSTRUCTIONS Before Assemb ly NO TE: Carefully inspect all car tons for shipping damage bef ore unpacking. If damage is found, contact y our shipper or T anita Cor poration immediately . Claims m ust be filed with the shipper as soon as possible after receipt of the damaged package .
Attaching Contr ol P anel Bo x to Scale 1. Find the control panel bo x, transf or mer brack et, C-clamp and f our (4) locknuts . 2. Attach the control panel bo x to the center handlebar r ail using the C-clamp . Also mount the transf or mer brac ket on two of the studs e xtending through the C-clamp .
Special Features T anita Cor poration’ s electronic scales come standard with these special f eatures: 1. A utozero: The scale will automatically displa y “0” when tur ned on. If a foreign object is placed on the platf orm bef ore tur ning the scale on, it will automatically zero out that object’ s weight.
T aking a W eight Measurement 1. Once the chair weight has been zeroed out and the displa y reads “0, ” ha v e the patient sit on the chair . 2. The moving dashes will loc k in and display the patient’ s weight. 3. W eight measurement should automatically be displa yed on the control panel bo x console.
7. Remov e weight and depress Zero s witch. W ait for a displa y reading of “0. ” Again add kno wn weight to platf or m. If reading is correct, no fur ther adjustments are necessar y . If reading is incorrect, repeat steps 6 and 6 until correct reading is obtained.
Qty Part Description Location 1 RES. MO X .47 1W 5% R4 1 RES. CF 2.7 1/4W 5% R2 1 RES. CF 10 1/4W 5% R12 1 RES. CF 22 1/4W 5% R7 1 RES. CF 150 1/4W 5% R29 1 RES. CF 120 1/4W 5% R6 2 RES. CF 560 1/4W 5% R13 R14 4 RES. CF 1K 1/4W 5% R3 R5 R15 R50 1 RES.
W ARRANTY T anita Cor poration (“T anita”) warrants that f or a period of three years from the date of purchase, this product will be free from def ects in material and workmanship. This w arr anty cov ers all components of this scale, with the e xception of the AC adapter .
An important point after buying a device Tanita 4800 (or even before the purchase) is to read its user manual. We should do this for several simple reasons:
If you have not bought Tanita 4800 yet, this is a good time to familiarize yourself with the basic data on the product. First of all view first pages of the manual, you can find above. You should find there the most important technical data Tanita 4800 - thus you can check whether the hardware meets your expectations. When delving into next pages of the user manual, Tanita 4800 you will learn all the available features of the product, as well as information on its operation. The information that you get Tanita 4800 will certainly help you make a decision on the purchase.
If you already are a holder of Tanita 4800, but have not read the manual yet, you should do it for the reasons described above. You will learn then if you properly used the available features, and whether you have not made any mistakes, which can shorten the lifetime Tanita 4800.
However, one of the most important roles played by the user manual is to help in solving problems with Tanita 4800. Almost always you will find there Troubleshooting, which are the most frequently occurring failures and malfunctions of the device Tanita 4800 along with tips on how to solve them. Even if you fail to solve the problem, the manual will show you a further procedure – contact to the customer service center or the nearest service center