Instruction/ maintenance manual of the product 3500 Tanita
Go to page of 8
Z E R O P O U N D S K I L O S O N O F F P o u n d s K i l o s www.tanita.com P u s h “ O N ” b u t t o n a n d w a i t u n t i l z e r o a p p e a r s b e f o r e s t e p p i n g o n s c a l e W E I G H T Model 3500 Stand-on Scale Cautions and W arnings .
T anita Cor poration’ s line of electronic scales uses the most sophisticated microprocessor technology av ailab le. Each precision instrument is designed to provide accurate , reliable and repeatab le weight measurements, as w ell as many other f eatures that mak e the weighing process simple, f ast and con v enient.
ASSEMBL Y INSTRUCTIONS Before Assemb ly NO TE: Carefully inspect the car ton for shipping damage bef ore unpacking. If damage is f ound, contact your shipper or T anita Corporation immediately . Claims must be filed with the shipper as soon as possible after receipt of the damaged package .
Attach Contr ol Bo x Assembly to Scale 1. Find the control panel bo x, transf or mer brack et, C-clamp and 4 lockn uts. 2. Attach the control panel bo x to the center handlebar r ail using the C-clamp . Also mount the transformer brac ket on two of the studs e xtending through the C-clamp .
Special Features T anita Corporation’ s electronic scales come standard with these special f eatures: 1. A utozero: The scale will automatically displa y “0” when tur ned on. If a f oreign object is placed on the platform bef ore tur ning the scale on, it will automatically zero out that object’ s weight.
Calibration Note: Y our scale has been accurately calibrated at the f actor y; howe ver , there is alwa ys a possibility that calibration has changed due to jarring in shipping. The f ollowing procedure should be f ollowed to chec k and/or correct any error in calibration.
Qty Part Description Location 1 RES. MO X .47 1W 5% R4 1 RES. CF 2.7 1/4W 5% R2 1 RES. CF 10 1/4W 5% R12 1 RES. CF 22 1/4W 5% R7 1 RES. CF 150 1/4W 5% R29 1 RES. CF 120 1/4W 5% R6 2 RES. CF 560 1/4W 5% R13 R14 4 RES. CF 1K 1/4W 5% R3 R5 R15 R50 1 RES.
W ARRANTY T anita Cor poration (“T anita”) warrants that f or a period of three years from the date of purchase, this product will be free from def ects in material and workmanship. This w arranty co vers all components of this scale , with the e xception of the AC adapter .
An important point after buying a device Tanita 3500 (or even before the purchase) is to read its user manual. We should do this for several simple reasons:
If you have not bought Tanita 3500 yet, this is a good time to familiarize yourself with the basic data on the product. First of all view first pages of the manual, you can find above. You should find there the most important technical data Tanita 3500 - thus you can check whether the hardware meets your expectations. When delving into next pages of the user manual, Tanita 3500 you will learn all the available features of the product, as well as information on its operation. The information that you get Tanita 3500 will certainly help you make a decision on the purchase.
If you already are a holder of Tanita 3500, but have not read the manual yet, you should do it for the reasons described above. You will learn then if you properly used the available features, and whether you have not made any mistakes, which can shorten the lifetime Tanita 3500.
However, one of the most important roles played by the user manual is to help in solving problems with Tanita 3500. Almost always you will find there Troubleshooting, which are the most frequently occurring failures and malfunctions of the device Tanita 3500 along with tips on how to solve them. Even if you fail to solve the problem, the manual will show you a further procedure – contact to the customer service center or the nearest service center