Instruction/ maintenance manual of the product SCC-TC02 National Instruments
Go to page of 16
USER GUIDE SCC-TC Series Thermocouple Input Modules The SCC-TC Series thermocouple input modules, SC C-TC01 and SCC-TC02, accept input signals from B-, E-, J-, K-, N-, R-, S-, and T-type thermocouples. Each module contains thermistor circ uitry po wer ed by 2.
SCC-TC Series Thermocouple Input Modules User Guide 2 ni.com bold Bold text denotes items that you must select in the software, such as menu items and dialog box options. Bold text also denotes parameter nam es. italic Italic text denotes variables, emphasis , a cross-reference, or an introduction to a key concept.
© National Instruments Corpor ation 3 SCC-TC Series Th ermocouple Input Mo dules User Guide ❑ Documentation – One of the following: • SC-2345/2350 User Manual • SCC-68 User Guide – SCC-TC S.
SCC-TC Series Thermocouple Input Modules User Guide 4 ni.com the inputs of the second-stage SCC. An example of dual-stage condit ioning in an SCC-TC0 X followed by an SCC-LP01 lowpass filter module. Sockets J9 to J16 also are av ailable for digital input/output (DIO) conditioning or control .
© National Instruments Corpor ation 5 SCC-TC Series Th ermocouple Input Mo dules User Guide Figure 1. SCC-TC0 X Signal Connections For more information about h ow to conf igure the SCC-TC0 X module usin g NI-D A Qmx, refer to the SCC Quick Start Guid e.
SCC-TC Series Thermocouple Input Modules User Guide 6 ni.com temperature measurements. T o make this con version, complete the following steps: 1. Measure the thermocouple v oltage. a. Read the thermocouple channe l on the E/M Series D A Q device V ESERIES [CH( X )].
© National Instruments Corpor ation 7 SCC-TC Series Th ermocouple Input Mo dules User Guide Detecting Open Thermocouples The SCC-TC0 X contains a 10 M Ω pul l-up resistor that connects to + 5 V to detect open thermocouples. To determine if you have an open thermocouple, check whether the co rresponding E/M Series DA Q device channel is saturated.
SCC-TC Series Thermocouple Input Modules User Guide 8 ni.com Cold-Junction Sensor The cold-junction sensor vol tage output varies from 1.91 to 0.58 V over a 0 to 55 ° C temperature range. Note NI-D A Qmx includes thermist or conv ersion utilities that implement the equations listed below .
© National Instruments Corpor ation 9 SCC-TC Series Th ermocouple Input Mo dules User Guide Calibrating System Offsets To calibrate the offset voltage of the SCC-TC0 X in the system using the E/M Series DAQ device, complete th e following steps. The E/M Series DAQ device must be powered for 30 minutes before you calibrate the offset voltage.
SCC-TC Series Thermocouple Input Modules User Guide 10 ni.com Specifications These specifications are typical at 25 °C unless otherwise stated. Analog Input Number of input channels.......................1 DIFF Input signals ..................... ...
© National Instruments Corpor ation 11 SCC-TC Series Thermocouple Input Modules U ser Guide Measurement Accuracy 1 1 T otal system measurement error fo r operating temper ature within ± 5 ° C of calibration temperature. Incl udes PCI/A T-MIO- 16XE-50 one-year accuracy specification of 0.
SCC-TC Series Thermocouple Input Modules User Guide 12 ni.com Cold-Junction Sensor Cold-junction sensor accuracy .......... ......± 0.4 ° C max from 15 to 35 ° C, ± 0.75 ° C max from 0 to 15 ° C and 35 to 55 ° C Output ............. ...........
© National Instruments Corpor ation 13 SCC-TC Series Thermocouple Input Modules U ser Guide Figure 3. SCC-TC02 Dimensions Weight TC-01 ................. ........... .............. .... 25 g (0.9 oz) TC-02 ................. ........... .............. .
SCC-TC Series Thermocouple Input Modules User Guide 14 ni.com Environmental Operating temperature ................ ............0 to 50 °C Storage temperature ............ .............. ......–20 to 70 °C Humidity ............ .............. .....
© National Instruments Corpor ation 15 SCC-TC Series Thermocouple Input Modules U ser Guide CE Compliance This product meets the essential re quirements of applicable European Directives, as amended .
An important point after buying a device National Instruments SCC-TC02 (or even before the purchase) is to read its user manual. We should do this for several simple reasons:
If you have not bought National Instruments SCC-TC02 yet, this is a good time to familiarize yourself with the basic data on the product. First of all view first pages of the manual, you can find above. You should find there the most important technical data National Instruments SCC-TC02 - thus you can check whether the hardware meets your expectations. When delving into next pages of the user manual, National Instruments SCC-TC02 you will learn all the available features of the product, as well as information on its operation. The information that you get National Instruments SCC-TC02 will certainly help you make a decision on the purchase.
If you already are a holder of National Instruments SCC-TC02, but have not read the manual yet, you should do it for the reasons described above. You will learn then if you properly used the available features, and whether you have not made any mistakes, which can shorten the lifetime National Instruments SCC-TC02.
However, one of the most important roles played by the user manual is to help in solving problems with National Instruments SCC-TC02. Almost always you will find there Troubleshooting, which are the most frequently occurring failures and malfunctions of the device National Instruments SCC-TC02 along with tips on how to solve them. Even if you fail to solve the problem, the manual will show you a further procedure – contact to the customer service center or the nearest service center